The data Toolbar contains all tools you can use on the image data.
Depending on the size of the window, some tools may not appear directly. When available, use the bottom arrows next to each toolbar to use the hidden tools.
The default tool used to select or edit added elements on the images.
Grab the image with this tool and move it in any direction to set a new view.
- Save Image to disk
Export and save the image on the computer file system. Different export options are available.
- Copy Image to clipboard
The displayed image is copied in the clipboard, and can be pasted in another application (Word, Excel, etc…). If Crop image is active, only the interesting area will be copied, otherwise everything.
RF Tools and Tracks¶
The RF Tool/sequence positions should be correct by default, as defined in the Plate layout parameters. However, you may have to manually adjust it in some specific situations, for example when the plate wasn’t absolutely well positioned in the Visualizer instrument.
The color of the RF Tool, track sequence, annotations, tips and exposure normalization is set either to yellow, red or black, depending on the image illumination, in order appear clearly by default. You can change the color by right-clicking on the corresponding tool and select the appropriate contextual menu entry.
- RF Tools
Toggle the visibility of the RF line and/or RF scale. Use the horizontal dotted lines to change the application and the front positions. Custom horizontal RF lines can be defined and saved to mark a specific position.
- Reset RF Tool positions
Reset the current RF Tool to its original position defined in the Plate layout parameters in Chromatography.
- RF Tip
Mark a spot on the plate with the exact RF or metric value.
Toggle the visibility of the track sequence to see where exactly does visionCATS “see” the tracks. Click on a track and move the vertical dotted lines in order to set the width of the tracks:
- Adjust Sequence position
Use the vertical dotted line on the left to set the position of the first track, and the vertical dotted line on the right to set the distance between tracks.
- Reset position
Reset the sequence position to its original values.
- Export to comparison
Extract the selected track(s) and paste them into the comparison tool (see Export to comparison for more details)
- Generate Profiles
Generates the profiles of all the (non-clean) images of the current analysis based on the current position of the RF Tool.
Deletes all output data (peaks, assignments, calibration functions and results) for evaluations based on a Visualizer step. Only available if the current analysis does not contain any locked evaluation based on a Visualizer step.
Turn automatic labelling for the tracks on or off. Select a combination of Vial ID, description, application volume or sequence number (i.e. track number) to be displayed.
Always hidden in the toolbar overflow:
RF tool color
Labels text size
- Spot Amp
Amplify a region on the plate (increase contrast of an averaged small area on plate).
- White Balance
Select a small area on the image and adjust the color of the whole image to make this area exactly white.
- Reset changes
Reset the Image settings adjustments (Spot Amp, Parameters and White Balance).
- Crop image
Remove or show the 5mm background around each plate image.
- Clean Plate Correction
Clean plate correction lets you subtract an image of the clean plate from images after chromatography. There is already some structure on the plate before chromatography (like background noise in an audio recording). The subtraction will result in a cleaner, sharpened image of the plate.
The following tools are only available for HDRI images (which are only available for R366).
- Normalize exposure
The saturation of the defined zone (exposure range) is normalized to 85%. Only a standard-dynamic-range around the wanted exposure will be displayed, all the data outside the range will be overexposed (white) or underexposed (black). If the exposure range is placed on a representative track of the plate (normally a reference), the important information of the image will be correctly displayed.
Enabling normalization will disable Clarify.
By default, the normalization will be enabled, regardless of the saved state. To change this behavior, see Global Configuration.
- Show exposure range
The current exposure range is displayed and can be modified graphically.
- Reset exposure range
The exposure range is reset to the original value, which is the first track of the plate.
Virtually change the illumination setting after capturing. Makes weak spots visible. It is a tone mapping technique which will compress the dynamic-range to makes most information visible on a limited dynamic-range monitor. The compression result in different color saturation and contrast than the reality, but more information is visible.
Enhancement slider: how much compression of dynamic-range will be done, making more or less information visible
Create a text box to put a comment on the plate image.
Indicate a specific area on the plate image with an arrow.
Delete the selected Annotation items.
Copy the selected Annotation items.
Paste the selected Annotation items.
Always hidden in the toolbar overflow:
Change the color of selected Annotation items.
Set text weight of selected Annotation items to bold.
Set text weight of selected Annotation items to italic.
Set text weight of selected Annotation items to underline.
- Rotate Left
Rotate selected Annotation items counter clock-wise.
- Right Rotate
Rotate selected Annotation items clock-wise.
Set text font of selected Annotation items.
- Text size
Set text size of selected Annotation items.