In the analysis view, the runs are added to a sequence of an HPTLC PRO SYSTEM. The runs placed in the same sequence are compatible in term of consumables and there are up to 5 runs pro sequence (if there is enough place for all vials in the rack). Then the HPTLC PRO SYSTEM supervisor will start a sequence in the HPTLC PRO SYSTEM Details view. If the HPTLC PRO SYSTEM has an HPTLC PRO Module PLATE STORAGE, he will have to place the carriers at their right positions into it. He will have to prepare consumables (see Consumables check) before he can start the execution.
An HPTLC PRO SYSTEM will handle alone up to 5 runs (5 different analyses) automatically. If the HPTLC PRO SYSTEM does not have a HPTLC PRO Module PLATE STORAGE, carriers will be manually loaded/unloaded using carrier manual loading/unloading positions.